PSN-L Email List Message
Subject: Re: Noise that my geophone picks up
From: "David Saum" DSaum@............
Date: Sat, 21 Jan 2006 10:23:48 -0500 (EST)
I have a similar noise problem with my 4.5 hz geophone
system that is resting on my basement office slab about
20m from a major highway.
I am using Amaseis freeware for data logging and display,
and I generally set the helicorder display low-pass filter
to 1 Hz, and that gets rid of a lot of the local noise.
When I receive and email notice of a teleseismic event,
I look at the data segment even if it does not show
up on my helicorder display. Often I can FFT the data
segment, see the quake spectral peak, and apply a bandpass
filter to bring the quake out.
One test to see if your electronics is working well is to
see if you can pick up the ~0.2 Hz microseism peak on
your geophone system. My system can generally pick
this up since I started using an AD8626 autozero opamp,
a pcb with star ground configuration, and a shielded
enclosure for my electronics and geophone.
Public Seismic Network Mailing List (PSN-L)
[ Top ]
[ Back ]
[ Home Page ]