Seislog expects to get an exact number of samples in a one second package, let's say 50. The SARA digitizer card mostly put out 50 but now and then maybe 49 or 51. Seislog (for windows) throws out the packets that don't have the exact sample rate. So what can you do?????
You can use the SARA acquisition software SeismowinPro. You can use the Seislog for Linux which is smarter than the Seislog for windows. It does not have the fancy GUI that other have but then it works real well. You can use SeisComP from GFZ also for linux.
Seismowin will make PSN, GSE and Seisan files it also has a LISS server so you can connect with earthworm systems. And you can use the Seisgram2k to look at the traces.
Well I ramble, there are many options.
If I can help just write a note.
Wednesday, April 18, 2007, 4:58:40 PM, you wrote:
Is anybody out there using "Mauro Mariotti, SADC" interface in conjunction with "Seislog" ?
I am experimenting with this setup and have an error message "Data has been lost" this message just rolling up the screen, Everything else seems to work ok, events detected and stopped etc but the events are dumped.
Many thanks in advance for any help.