PSN-L Email List Message
Subject: Re: Tiltmeter as Horizontal Seismometer
From: Pete Rowe ptrowe@.........
Date: Sat, 15 Sep 2007 07:48:27 -0700 (PDT)
Hi again Chuck
I left out one important piece of the circuit. Chris'
reply to you jogged my memory. Since you drive the
bridge with AC, you have to have a synchronous
rectifier after the gain stages to turn the signal
back to DC. This is done with an analog switch and an
op amp. Easy.
As Chris mentioned, the homemade liquid filled tube
may be your best bet for building the sensor yourself.
I saw one of these at Cal Tech many years ago. It was
about 2 meters long and had a horizontal tube filled
with Mercury. They sensed the level of the Mercury in
a cup at each end capacitively. With today's
capacitive sensor ICs, this should be easy to do.
--- Chuck Burch wrote:
> I am drafting plans to build a three electrode
> spirit level style tiltmeter to use as a
> horizontal seismometer. Any hints on construction?
> What liquid is best? Are
> surface tension and/or viscosity critical?
> I'm planning on using a Wheatstone bridge detector
> circuit. Can this be improved?
> Thanks for your comments.
> Chuck I. Burch
> Public Seismic Network Mailing List (PSN-L)
> To leave this list email
> PSN-L-REQUEST@.............. with
> the body of the message (first line only):
> See http://www.seismicnet.com/maillist.html for more
Be a better Heartthrob. Get better relationship answers from someone who knows. Yahoo! Answers - Check it out.
Public Seismic Network Mailing List (PSN-L)
[ Top ]
[ Back ]
[ Home Page ]