PSN-L Email List Message
From: Steinar Midtskogen steinar@.............
Date: Tue, 29 Jul 2008 20:42:56 +0200
I've had a SEP seismometer running for a month or so and everything
has been fine. However, a couple of days ago while we were away on
vacation it began reporting no activity. When I got back home I found
that the mass had moved all the way to the side, so I had to change
the tilt a little so it would move back to the centre again.
Is this normal behaviour for a Lehmann seismometer? The mass moved
away from the center at a time the house was empty, so I can't really
think of any external disturbances which could have caused this.
Public Seismic Network Mailing List (PSN-L)
[ Top ]
[ Back ]
[ Home Page ]