PSN-L Email List Message
Subject: Re: Strong Damping Magnets and pick up coil
From: "Geoffrey" gmvoeth@...........
Date: Wed, 4 Mar 2009 11:13:25 -0700
Hello Mr. Green;
I assume VS stands for vertical Short Period.
Id like to put in my 2 cents for your thoughts.
No one as of yet seems to have built what
I consider to be a correct sensor.
You really need a two sensor device that is grounded in
the middle to be a single three wire device.
This allows to double the signal level as well as provide
a proper balanced input into a differential amplifier.
Both sensors need to be identically matched so it
might be difficult to build.
You cant just make a centered tapped coil.
It needs to be two identical coils arranged in
a series additive fashion feeding the common center
point and the plus minus signals on a 100%
Shielded Underground cable.
The magnets need to be matched as well in strength.
Just thought Id like to say this since everyone is going
for only a half kinda sensor and not what the input amplifier design
sheets seem to require. ( two identical symmetrical inputs grounded in the center ).
This makes construction more difficult if it is one of a kind.
----- Original Message -----
From: "Robert O. Green"
Sent: Wednesday, March 04, 2009 7:45 AM
Subject: Strong Damping Magnets and pick up coil
>I am working on a VS system with Larry's coil, magnet and signal-chain.
> Is there any impact to the pick up coil signal from the strong magnets
> mounted nearby?
> Public Seismic Network Mailing List (PSN-L)
> To leave this list email PSN-L-REQUEST@.............. with
> the body of the message (first line only): unsubscribe
> See http://www.seismicnet.com/maillist.html for more information.
Public Seismic Network Mailing List (PSN-L)
[ Top ]
[ Back ]
[ Home Page ]