PSN-L Email List Message
Subject: Re: No data from the lehman seismomter
From: =?ISO-8859-1?Q?J=F3n_Fr=EDmann?= jonfr@.........
Date: Mon, 04 May 2009 23:15:29 +0000
This is not the cable. This appears to be the coil. I have tried to fix
it, but it doesn't appear to work.
This happens when I just got my lehman seismomter to work as it should
have. It now has a period of ~5 seconds with out special setup (0.2Hz).
I don't know what did happen, this just stopped working. :-(
On m=E1n, 2009-05-04 at 16:34 -0600, tchannel wrote:
> Hi Jon, I have a Lehman which gave me the same problem, it was a bad=20
> connector, it simply was not making contact.
> ----- Original Message -----=20
> From: "J=F3n Fr=EDmann"
> To: "PSN-Postlist"
> Sent: Monday, May 04, 2009 3:45 PM
> Subject: No data from the lehman seismomter
> Hi all
> For some reason, the coil did go silent on me. I have no idea what is
> wrong, I have tried to change about wire and other things that might
> create this issue.
> But nothing has worked, and I am out of ideas. The boom swings correctly
> and everything else is working as it should be, but no data from the
Public Seismic Network Mailing List (PSN-L)
[ Top ]
[ Back ]
[ Home Page ]