PSN-L Email List Message
Subject: Re: Did I get 12 second period ?
From: =?ISO-8859-1?Q?J=F3n_Fr=EDmann?= jonfr@.........
Date: Sat, 30 May 2009 00:56:56 +0000
Here is a signal that I did get tonight with a 12 second period lehman
type sensor. I have changed the setup a bit from previus setup, this
appears to give me a clear signal. The magnets are now aligned with the
coil, so the magnetic field is now stable when the coil moves. That way
I got rid of increse, decrese effect of the magent in regards to the
coil. The setup is not complete, as I need correct magnets. But I have
setup a test magnets for the time being. The first resaults are good.
The sensor has a high peak around 12 seconds, but is able to detect
signal up to 0.1Hz, as can been seen on this earthquake detection. It
goes as low as 20 seconds I think.=20
This is work in progress, I am going to get rare-earth magnets next
month to make the final improvement to my lehman seismomter.
Here is the signal. This earthquake was Mw4,92 in size at the distance
of 180 km.
Public Seismic Network Mailing List (PSN-L)
[ Top ]
[ Back ]
[ Home Page ]